Bacterial spores which tends to be the most persistent, can
survive for long periods in adverse conditions.
Steam is provided to the chamber from external source e.g.,
All the processes of sterilization, vacuum and drying
are automatic. At the end of operation, a buzzer is
sounded and the pressure / steam is automatically turned
Air in chamber is completely removed. The sterilization
cycle timer starts to count only after set pressure is
achieved, which ensures complete sterilization.
After sterilization, heating/vacuum operation
automatically ensures rapid and complete drying of
Pressure &Temperature Control
A high performance pressure regulator and thermostat are
fitted to maintain the required temperature and pressure
inside the chamber. If an abnormal pressure is produced,
pressure switch and safety valve go into operation.
Doors are locked automatically and cannot be opened
until pressure is exhausted. However, manual operation
is also available.
Digital Timers for sterilization and drying cycle.
Auto Releaseof Air & Steam
Automatic release of air from chamber before the start
of sterilization cycle and auto release of steam at the
end of cycle.